View and Download Invacare Platinum 5 service manual online. Invacare Platinum Series Oxygen Concentrator. Platinum 5 Oxygen Equipment pdf. 35 5-Liter Models. Respironics Millennium 5 Oxygen Concentrator. The Respironics Millenium Oxygen Concentrator is one of the most popular oxygen generators available today. It features a maximum liter flow of 5 liters per minute. The used oxygen concentrator has been refurbished by our factory trained and authorized service technicians and are certified to operate to manufacturers specifications. View and Download Invacare Platinum 5 service manual online. Invacare Platinum Series Oxygen Concentrator. Platinum 5 Oxygen Equipment pdf manual download. Also for: Platinum 10, Platinum xl.
• ® iter ompaCt xygen onCentrator Service Manual Danger – no SMoking For Product Numbers: 525DS 525KS CAUTION-Federal (U.S.A.) law restricts this device to sale by or on the order of a physician. • T a B l e o F c o n T e n T S general inForMaTion Introduction..... 3 Important Safeguards....3 Safety Precautions and General Warnings....3 unPacking anD SeTuP Initial Inspection.....4 Patient Setup.....4 Operating Instructions..... • Respiratory Technical Service Department at 1-800-338-1988 (814-443-4881). NoTe– DeVilbiss reserves the right to alter or change the design of the DeVilbiss Oxygen Concentrator series.
Hence, slight differences in construction or components may exist between the unit in hand and what is described in this manual. • It is suggested that an initial inspection be performed upon receiving the oxygen concentrator. After removing the DeVilbiss Oxygen Concentrator from the carton, examine it for any external damage. If shipping damage has occurred, contact the DeVilbiss Customer Service Department at 1-800-338-1988 (814-443-4881) for specific instructions. Save the carton for possible later return.
• Remove the power cord completely from the strap. Make sure the power switch is in the “Off” position. 115 volt units– Insert the plug into an electrical outlet. The DeVilbiss Oxygen Concentrator uses a two-prong polarized plug and is double- insulated to protect against electric shock. • Do not attempt to operate the unit without the air filter or while the filter is still damp. NoTe– T heairfiltershouldbemonitoredmorecloselyinenvironmentswithabnormalamountsofdustandlint. CauTion– OperationoftheDeVilbissOxygenConcentratorinextremeenvironmentsorwithouttheairfilterwillprematurelyoccludethe intakebacteriafilterandcauseadecreaseintheunitperformance.
Exterior cabinet The patient should clean the concentrator exterior cabinet by using a damp cloth or sponge with a mild household cleaner and wiping it dry. • PerioDic HoMecare ProviDer PrevenTaTive MainTenance Every DeVilbiss Oxygen Concentrator is tested at the factory.
To assure continued trouble-free performance, the following preventative maintenance should be performed by the homecare provider during periodic oxygen patient visits. Failure to properly maintain the unit will void the warranty. Stihl St400 Parts Manual. • The DeVilbiss Oxygen Concentrator operates on a timed cycle. The cycling is controlled by the PC board. The PC board will send voltage to the valve causing it to shift and alternately pressurize the sieve beds. • T r o u B l e S H o o T i n g SiMPliFieD TrouBleSHooTing The key to simple troubleshooting is to recognize which type of problem exists and select the most effective approach to solving the problem. The different types of problems and the approaches for solutions are as follows: Type i—The unit runs but a low pressure and flow or high pressure condition exists.
• T r o u B l e S H o o T i n g TrouBleSHooTing cHarT a visible alarm audible alarm compressor Power light other Symptoms Possible cause Possible remedy Pulsating air noise Intake filter not in place or defective Check filter and replace if necessary Compressor intake hose disconnected Reconnect hose. • T r o u B l e S H o o T i n g TrouBleSHooTing cHarT B visible alarm audible alarm compressor Power light Blinking Pulsing other Symptoms Possible cause Possible remedy Fan off Line cord not properly installed or defective Insert plug in receptacle or replace line cord. • T r o u B l e S H o o T i n g TrouBleSHooTing cHarT e visible alarm audible alarm compressor Power light other Symptoms Possible cause Possible remedy Fluctuating or no flow System pressure below 9 psi (62.1 kPa) due to: 1.